RTD™ Wound Dressing is made of polyurethane/ polyether foam, Methylene Blue (0.25 mg/g), Gentian Violet (0.25 mg/g), and Silver Ion (7 mg/g).
The RTD™ Wound Care Dressing is indicated for the treatment of partial to full thickness wounds with moderate to heavy exudate, including pressure ulcers, venous stasis and arterial ulcers, diabetic foot ulcers, skin tears, graft wounds and donor sites, surgical/post-operative wounds, first and second degree burns, lacerations and abrasions and any other wound inflicted by trauma.
1. Measure and Cut RTD™ to size of wound.
2. Apply RTD™ dressing directly to wound.
3. Secure with appropriate secondary dressing.
-RTD™ can be rolled and packed into the wound.
-For best results, ensure RTD™ is in direct contact with the wound.
-Suggested RTD™ dressing change interval is every 3 days, or as needed.
RTD™ Wound Care Dressings are not indicated for third-degree burns.
Do not use RTD™ Wound Care Dressing on patients with known sensitivity to silver.
In the event of clinical infection, RTD™ Wound Care Dressing is not intended to replace systemic therapy or appropriate treatment protocol for infection.
Latex Free-Not made with natural rubber latex
Caution: Rx only-Federal Laws restrict this device to sale by or on the order of a licensed physician.
Clean the wound with normal saline or other appropriate cleaning agent per facility protocol.
Standard precautions should be used to prevent spread of infectious agents when dressing wounds.
Securing RTD™ Wound Dressing in Place: Use gauze, Kerlix™, Omnifix™, Mefix™, other non-occlusive adhesives or medical grade tape.
RTD™ Wound Dressing may be left on for a maximum of three days. Remove the dressing sooner if foam becomes saturated. If RTD™ is difficult to remove, soak with saline or sterile water. A meshed non-adherent layer is recommended to prevent sticking or ingrowth at the later stages of wound healing.
Dressing should be in direct contact with the wound for best results. Fit to edges of the wound. For moderate to heavy exudate wounds and initial dressing applications, the 1/4” thickness is recommended.
The 1/8” thickness is recommended for lighter exudate wounds. If the dressing becomes saturated before the desired dressing change, a second dressing may be place on top for continued absorption and vertical wicking.
Considerations for Tunneling Wounds: Cut the dressing (1/4” thickness) into strips or cut and roll to appropriate size. Leave a “tail” at the end of the dressing to assist with easy removal. Insert into wound tunnel. Do not allow dressing to become fully saturated.
Considerations for Ostomy Dressing:
Trim the outer diameter of the RTD™ Dressing (1/8” thickness) to fit just inside the adhesive rim of the pouch. Trim the inner diameter of the dressing to fit the stoma. If a peristomal wound is present, cut the 1/8 inch thickness RTD™ Wound Dressing to the wound size and placed on the wound. Secure the pouch wafer using preferred tube paste around the precut opening for the stoma.
Considerations for Skin Graft and Donor sites:
To apply the RTD™ Wound Dressing after graft placement. Start with meshed non-adherence dressing such as a petroleum impregnated, or a silicone based wound contact layer. Cut the non-adherent so that it extends one centimeter beyond the edge of the skin graft.
Cut the RTD™ Wound Dressing the same size and place over the non-adherent. Secure dressing per standard protocol. Leave in place per doctor’s orders (usually 4 to 7 days).
Considerations for around G-Tubes, Chest Tubes, Tracheostomy Tubes, Pic Line or Central Venous Lines:
Cleanse the area per facility protocol. Cut the 1/8” RTD™ Wound Dressing with a slit and slight cut out to allow it to fit securely around the tubing and lie flat against the adjacent skin. For the tracheostomy tube, allow the dressing to fit the foam underneath the flange as able. Secure with a non-occlusive adhesive or medical grade tape. Leave in place for up to three days.
Considerations for use Under Negative Pressure Therapy (NPWT) Device:
Cut the RTD™ Wound Dressing to size. Fenestrate the foam. Place the RTD™ Dressing in contact with the wound bed. Place the NPWT foam on top of the RTD™ Dressing and seal the wound per the NPWT Device protocol.
RTD™ can be used under compression wrap or occlusive adhesive with medical supervision or trained care givers.